F245 Axial Compensated Loadcell
Standard Ranges 100, 200, 400 and 800N (10 to 80kgf)
- High accuracy
- Compact axial geometry
- Misalignment error compensation
- Standard 1 year warranty
- Low deflection at full load
- Traceable calibration with certificate included in the standard price
Non-linearity - Terminal
Creep - 20 minutes
Rated output - Nominal
Rated output - Rationalised
Rationalisation tolerance (applies to single direction calibrations)
Zero load output
Temperature effect on rated output per Â°C
Temperature effect on zero load output per Â°C
Temperature range - Compensated
-10 to +50
Temperature range - Safe
-10 to +80
Excitation voltage - Recommended
Excitation voltage - Maximum
Insulation resistance - Minimum at 50Vdc
Overload - Safe
Overload - Ultimate
Weight - Nominal (excluding cable)
40 to 50
Standard ranges up to 200N have an aluminium body and stainless steel case. Ranges above 200N have stainless steel bodies and cases.
Geometry: Flexure strain assembly in cylindrical housing, open or weather sealed with end internal fixing. For universal use in tension and compression, with compensation for off axis load inputs.
The F245 is ideally suited to low range engineering force measurements and process weighing. When precision and easy installation are required various configurations allow the loadcell to be used in both tensile and compressive applications.
We are happy to design variants of this loadcell to meet your specific requirements. Versions can be manufactured for higher temperature operation. Please consult our engineering department.
Change the first 0 to an R for the IP65 version.
Structural Stiffness - Nominal
3.2 x 106
6.4 x 106
3.0 x 106
6.0 x 106
- AL = Applied load.
- RL = Rated load.
- Temperature coefficients apply over the compensated range.
The loadcell is fitted with 2 metres of PVC insulated 4 core screened cable type 7-2-4C.
Excitation + = Red, Excitation - = Blue, Signal + = Yellow, Signal - = Green, Screen = Orange.
Reverse the signal connections to obtain a positive signal in tension mode. The screen is not connected to the loadcell body.